Sale ends soon!
Regulate preparation and Notice 1. Connection (please refer the product mix content and connection method above). 2. Regulate the voltage selection, 115V for the standard voltage range for the 90-120V, 230V for 210-240V range. (This is very important, a wrong choice is easy to damage or affect the effectiveness of the instrument) 3. Confirm the main power switch off position at 0. 4. Plug in the power and turn the main power switch. 5. Gently coated with special rubber on the treating parts. (Cleaning before conditioning, more better) 6. Installed the RF head on the probe handle. (Connect the metal cylinder into the connection hole to the deepest) 7. Subject’s hand with a non-drying (dry cleaning) grip loop handle, thumb put on the red power button on the safe preparation, if not press the red switch, the machine will not work. 8. Regulate the time required. Specifications Voltage & current: 90-120V/210-240V Frequency: 50-60HZ Supreme power consumption: 350W Function frequency: 3Mhz Material: ABS resin (host) NW: about 3.5KG Packing specification: 350MM*350MM*280MM GW: about 5KG